GRETAGMACBETH SPECTROLINO DRIVER DOWNLOAD
This Buy It Now listing has ended. Learn More – opens in a new window or tab Any international shipping is paid in part to Pitney Bowes Inc. Buyers can restrict the circle of potential business partners without extensive checks and favor companies that make honest and righteous shops with high probability. Power is distributed to the stepper motors, front panel circuit board and the SpectroLino from the main circuit board through various other cabling. Tried resetting the SpectroLino as outlined on page 17 of the User Guide. Email to friends Share on Facebook – opens in a new window or tab Share on Twitter – opens in a new window or tab Share on Pinterest – opens in a new window or tab.
|Date Added:||18 May 2010|
|File Size:||31.63 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Buyer complaints can lead to the withdrawal of the seal. This seller is currently away until Jan 02, The item may have some signs of cosmetic wear, but is fully operational and functions as intended.
Subject to credit approval. Your contact is typing. Interest will be charged to your account from the purchase date if the balance is not paid spectroluno full within 6 months.
GretagMacbeth SpectroScan White Densitometer Spectrolino | eBay
The Spectrolino has outstanding technical performance. Digging a little deeper, I removed the main, intermediate, and small circuit boards to study them more closely.
At this point I began thinking that the issue was with the SpectroScan table and not the SpectroLino, but I wanted confirmation that my SpectroLino was working. To do so, I connected the SpectroLino directly to the computer with the necessary cabling and hardware: However, after searching the internet for as long as I could handle, I was still empty-handed. Any international shipping is paid in part specrtolino Pitney Bowes Inc.
GretagMacbeth SpectroScan T x/y table Repair | Bodenzord
What I was looking for most specifically was the pin configuration of the SpectroLino, which I found on page Therefore, I was beginning to hypothesize that I had either a power or communications issue between gretagmwcbeth main circuit board and the SpectroLino. Email to friends Share on Facebook – opens in a new window or tab Share on Twitter – opens in a new window or tab Share on Pinterest – opens in a new window or tab Add to watch list.
Learn More – opens specfrolino a new window or tab. This is not the most elegant way to fix a problem. Machineseeker app for iPhone and Android.
In fact, they are no longer around because they were purchased and taken over by X-Rite many years ago, who also no longer supports the SpectroScan. He did say, however, that he would contact a colleague in the UK that may be of help.
Machineseeker Trust Seal
Although I had previously measured 24v on pin 7, I surprisingly discovered that I had no continuity on this pin. Let me describe the issue in more detail. Seller assumes all responsibility for this listing. I bought this unit used about eight years ago, and it has served me well throughout this time. Gretaymacbeth a valid country. Refer to eBay Return policy for more details.
This application communicates with the SpectroScan, collects the measurement readings, and saves them to a file that is ultimately used with ProfileMaker to create a printer profile. When I attempted to spectrolijo a target recently, MeasureTool gave me an error: We will honor a return if the item s have been deemed to be defective during the warranty period.
In my case, this occurred and appeared normal.
In order to return the item, you must email us first and obtain an RMA number for us to process. Please correct your entries in the highlighted boxes. I brought the computer into my electronics class to try and repair the garbled display. After testing the cables, the good news was that each circuit on both vretagmacbeth tested good for continuity.